What is Daily Calibration?
Daily calibration is a procedure that is designed for the instrument operator to bring the instrument to a known state utilizing a factory supplied reference standard of traceable values.
The procedure for daily calibration is simple and straightforward. Two measurements are taken with the instrument, one measurement of a ceramic white standard with traceable values, and one measurement of absolute black.
The measurement of the white ceramic standard must always be performed with the standard that was assigned to the respective instrument. Each white standard is serialized and therefore easily paired with the correct instrument. The absolute black measurement, referred to as a 'zero calibration' or 'black trap measurement' can be performed by utilizing a special attachment for the instrument which absorbs all light (black trap, or zero calibration box), or by taking a measurement with nothing in front of the instruments measurement port for at least 3ft (open port black calibration). In either scenario the end result is the same in that no light will be reflected back into the instrument and therefore the measurement result will be of an absolute black.
Both the white standard reading and the absolute black reading are utilized by the instruments onboard processor to set the instrument to a known state of calibration.
How often should daily calibration be performed by the operator of the instrument?
Daily calibration of color measurement devices should be performed in the same environment and conditions as the measurements with the device will be taken. The instrument and its calibration standard should be given sufficient time to acclimatize to the environment where calibration and measurement will occur. Typically with stationary instruments a minimum interval that daily calibration should be performed is every 8 hours or every shift. However there are various cases where daily calibration should be performed more often such as:
If you have any questions on calibration interval please consult the user's manual of the instrument or with Konica Minolta's team of application engineers, they can recommend the optimum calibration interval based on your specific measurement requirements. Keep in mind, daily calibration only takes a matter of seconds, so when in doubt, re-calibrate!
There are several important considerations when performing daily calibration of color measurement devices, they can be summarized as:
© 2006-2017 Konica Minolta Sensing Americas, Inc.