Daily calibration of color measurement devices should be performed in the same environment and conditions as the measurements with the device will be taken. The instrument and its calibration standard should be given sufficient time to acclimatize to the environment where calibration and measurement will occur. Typically with stationary instruments a minimum interval that daily calibration should be performed is every 8 hours or every shift. However there are various cases where daily calibration should be performed more often such as:
- If the environmental conditions of temperature and humidity change. The calibration process allows the operating environment to be taken into account to provide consistent results.
- If the measurement mode of the instrument is changed, for example specular component, reflectance, transmittance, etc...
- If the aperture size of the instrument is changed.
If you have any questions on calibration interval please consult the user's manual of the instrument or with Konica Minolta's team of application engineers, they can recommend the optimum calibration interval based on your specific measurement requirements. Keep in mind, daily calibration only takes a matter of seconds, so when in doubt, re-calibrate!
There are several important considerations when performing daily calibration of color measurement devices, they can be summarized as:
- Ensure the white calibration tile is free from scratches, dirt, or fingerprints. Any debris or abnormality can affect the calibration of the device in question.
- Ensure the white calibration tile is seated correctly to the instruments measurement aperture. Any angular misalignment can cause variation in the measurement of the calibration standard and introduce error.
- Ensure you are utilizing the correct white calibration standard for your device.
- For black calibration utilizing a zero calibration box ensure that the zero calibration box, or black trap, is free from dust and debris.
- For black calibration utilizing an open port calibration, ensure that there is no objects in front of the instruments measurement port less than three feet, and do not place the measurement port facing toward a light source.